We develop and produce innovative device technology that is made for daily use! Therefore, it is very important to us that you use it effectively and profitably. We want to support you in your everyday work with our measuring devices in the best possible way. Our menu navigation is simple and uniform, regardless of whether it is a gas detector or a pressure measuring system.
What is a tutorial?
To offer you even more assistance we have recorded videos, so-called tutorials. These are small “operating instructions” in a moving image format – for example in the form of a video or an animated graphic – and are intended to guide you to the solution of a possible problem.
Step by step
We explain our measuring devices or individual menu items in detail. The videos are adapted to our proven operation and our uniform handling, so that you can help yourself quickly and easily. Should there be any questions or uncertainties afterwards, our internal and external sales force for technology, support and advice is of course also available to you by phone at any time.
Support where and when you want it
Watch the videos when and where it is convenient for you, whether before using the devices or in between on the construction site. Our explanations and possible solutions lead directly to the desired goal. In addition, videos are more effective than text, because what you see remains longer in your memory than what you read.
Feel free to contact us and let us find a solution together.
In the menu point Bump Test, the device is checked for the following: Free flow of the gas paths in the measuring device, Reactivity of the measuring device to test gas, Intact sensors, alarms (optical, acoustic and vibration) First the zero point can be set. Then press Start. The device displays “Please add test gas”. In the case of the OLLI I apply the 5K gas (5 components gas). Test gas can 5 components 1,65 Ltr 35 bar 2.2 % CH4, 150 ppm CO, 2.5% CO2, 15% O2, 25 ppm H2S, rest N2 Capacity: 58 litres Only when the checkboxes are filled in, the bump test is passed and can be verified. The bump test can now be completed with Enter. Please remove test gas. Now the bump test is completed successfully.
Connect the Link via USB before starting the Esders Update Tool. Switch on the device. Establish the infrared connection between the measuring device and the Link. And press Start in the Update Tool. After the beep, the Bluetooth connection between the Link and the measuring device has been established. The measuring device switches to data mode. If a new firmware version is available, install it using the Update Tool. During the update process the link (our data module) communicates with the device via Bluetooth. No infrared communication is required. The infrared communication is only needed for the first coupling of the Link to the measuring device. The firmware has been updated. By restarting the device the firmware update is successfully completed.